Pharmaceutical Residual Solvent Analysis: A Comparison of GC-FID and SIFT-MS Performance

: Residual solvents in pharmaceutical excipients, active pharmaceutical ingredients (APIs), and ﬁnished products are usually analyzed using gas chromatography (GC)-based techniques according to a pharmacopeial monograph, such as the United States Pharmacopeia’s (USP) chapter <467>. GC analyses are often slow, which limits sample throughput. Selected ion ﬂow tube mass spectrometry (SIFT-MS) removes the rate-limiting chromatographic separation step, potentially of-fering faster sample analyses. This approach was demonstrated recently with the publication of an alternative SIFT-MS procedure which was successfully validated against the performance criteria in USP chapter <1467>. The present study expands upon the previous work by conducting a head-to-head comparison of GC-ﬂame ionization detection (GC-FID) and SIFT-MS procedures. The results obtained in this cross-platform study demonstrated similar performance for the GC-FID and SIFT-MS procedures for linearity (R 2 > 0.94 and 0.97, respectively) and repeatability (<17%RSD and <10%RSD). For accuracy and recovery, acceptance criteria (within 20%) were achieved for most compounds across the two drug products (SIFT-MS suffered fewer failures, possibly due to shorter wait times prior to analysis). Additionally, SIFT-MS analyzed samples over 11-fold faster than GC-FID, increasing daily sample throughput and reducing the time taken to determine the result. This study therefore suggests that residual solvent analysis using SIFT-MS may support workﬂow improvements for pharmaceutical manufacturers.


Introduction
Residual solvents in pharmaceutical products are defined by the International Council for Harmonization of Technical Requirements for Pharmaceuticals for Human Use (ICH) as "organic volatile chemicals that are used or produced in the manufacture of drug substances or excipients, or in the preparation of drug products. The solvents are not completely removed by practical manufacturing techniques" [1] (p. 1). Residual solvents are grouped into three classes based on toxicity. Class 1 solvents are known to have unacceptable toxicities and should be avoided (the exception is 1,1,1-trichloroethane, which is an environmental hazard). Solvents in Class 2 are associated with less severe toxicity but should be limited to protect patients. Finally, Class 3 solvents are those with low toxic potential and should be used where practical to do so. Note that Class 2 solvents are further divided into subclasses 2A, 2B, and 2C. This is based largely on the combination of the permitted daily exposure (PDE) determined from toxicological considerations (concentration in solution) and the degree to which they partition to headspace. From Class 2A to 2C, there is generally significant diminution in headspace partitioning, to the extent that the United States Pharmacopeia (USP) General Chapter <467> Residual Solvents [2] states that acceptance criteria are not met for several compounds in Class 2C, and an alternative procedure must be used. The present study addresses the need to compare the performance of the SIFT-MS alternative procedure [12] to the standard USP <467> procedure using GC-FID [2]. It does this by analyzing identically prepared samples on side-by-side SIFT-MS and GC-FID instruments. Specifically, comparison is made of the repeatability and linearity of the standard solutions, and the accuracy and recovery for spiked solutions prepared from two fin- The present study addresses the need to compare the performance of the SIFT-MS alternative procedure [12] to the standard USP <467> procedure using GC-FID [2]. It does this by analyzing identically prepared samples on side-by-side SIFT-MS and GC-FID instruments. Specifically, comparison is made of the repeatability and linearity of the standard solutions, and the accuracy and recovery for spiked solutions prepared from two finished acetaminophen (paracetamol) products-a tablet and an oral suspension. Results obtained for these formulations also enable several recommendations to be made when applying the SIFT-MS procedure to new products.

Automated SIFT-MS Analysis
The SIFT-MS analytical technique has been described in detail previously [10,14]. Briefly, SIFT-MS is a DIMS technique that analyzes air and headspace continuously by using ultra-soft chemical ionization that efficiently ionizes a very broad range of VOCs, but does not ionize the bulk constituents of air. A microwave discharge in air is used to generate the reagent ions, with eight available (H 3   and NO 3 − ) on the SIFT-MS instrument used in this study (Voice200ultra; Syft Technologies Limited, Christchurch, New Zealand) [15]. Rapid switching of reagent ions provides high specificity because the multiple reaction mechanisms give independent measurements of each analyte, while the absence of chromatographic separation means that it is straightforward to analyze VOCs of diverse chemical functionalities. Instrument detection limits in the part-per-trillion by volume (pptV) range are typically achieved for 1 s ion dwell times for direct analysis of air, with no preconcentration or drying required [16,17].
The SIFT-MS instrument was equipped with a GERSTEL multipurpose (MPS) autosampler (Robotic Pro; Mülheim, Germany). Samples were incubated in a virtual twelve-place GERSTEL agitator (composed of two physical six-place agitators) prior to sampling of the headspace and subsequent injection into the SIFT-MS instrument through a GERSTEL septumless sampling head. The GERSTEL Maestro software's "PrepAhead" sequence schedule is shown in Figure 2a.
The headspace conditions for all analyses used 6 mL of solution in a 20 mL headspace vial incubated at 60 • C for 45 min. A 2.5 mL aliquot of headspace was removed via a heated syringe (150 • C) and injected into the SIFT-MS instrument's sample inlet at 25 µL·s −1 , with a zero-air make-up gas flow through the heated inlet (150 • C) to ensure that the total flow into the instrument was 25 mL·min −1 . After the injection, the syringe was flushed with zero air for 1 min at 200 mL·min −1 . Figure 3 shows an example headspace injection for Class 2A solvents, where data for all reagent ions were collected in the same analytical run and concentrations are calculated by averaging data from 50 to 120 s. For residual solvents analysis, some compound-dependent signal drift is observed during sample injection, but this is consistent as demonstrated by the good repeatability (Table 1).

Automated GC-FID Analysis
GC-FID analysis was conducted using headspace operating parameter set 3 and procedure A of USP General Chapter <467> Residual Solvents [2]. The Agilent 7890 GC-FID (Agilent Technologies, Santa Clara, CA, USA) was coupled with a multipurpose autosampler (MPS Robotic Pro (dual head); GERSTEL, Mülheim, Germany). Samples were incubated for 45 min at 80 • C. The optimized schedule using the GERSTEL Maestro PrepAhead software is shown in Figure 2b.  The headspace conditions for all analyses used 6 mL of solution in a 20 mL headspace vial incubated at 60 °C for 45 min. A 2.5 mL aliquot of headspace was removed via a heated syringe (150 °C) and injected into the SIFT-MS instrument's sample inlet at 25 μL s −1 , with a zero-air make-up gas flow through the heated inlet (150 °C) to ensure that the total flow into the instrument was 25 mL min −1 . After the injection, the syringe was flushed with zero air for 1 min at 200 mL min −1 . Figure 3 shows an example headspace injection for Class 2A solvents, where data for all reagent ions were collected in the same analytical run and concentrations are calculated by averaging data from 50 to 120 s. For residual solvents A 1 mL aliquot of headspace was injected at 200 µL·s −1 into the split/splitless inlet (5:1 split ratio) of the GC. Separation was achieved using a VF-624ms column (30 m × 0.32 mm × 1.8 µm; Agilent Technologies, Santa Clara, CA, USA) with a helium flow rate of 2.15 mL·min −1 . The oven was initially held at 40 • C for 20 min before a ramp of 10 • C·min −1 to 240 • C, where it was held for 20 min. Eluting components were detected using a flame ionization detector (FID) operating at 250 • C with nitrogen makeup gas, hydrogen, and air flows of 25, 30, and 400 mL·min −1 , respectively. A representative chromatogram is shown in Figure 4.
analysis, some compound-dependent signal drift is observed during sample injection, but this is consistent as demonstrated by the good repeatability (Table 1).

Sample Preparation and Analysis
The study was conducted in two separate runs (Table S2) and different commercial standards were utilized for each. For Run 1, USP Class 2A and 2B standards were used (part numbers 1601281 and 1601292, respectively; USP, Rockville MD, USA), while for Run 2, Sigma-Aldrich Class 2A and 2B standards were employed (part numbers PHR1064 and PHR1065, respectively; Sigma-Aldrich, Gillingham, UK).
Two acetaminophen (paracetamol) formulations were selected for the accuracy and recovery exercise: tablets (500 mg per tablet; Co-op private label, UK) and an oral suspension (250 mg per 5 mL dose; Calpol Six-plus sugar-free, Johnson & Johnson, High Wycombe, UK).
Preparation of samples (combining Classes 2A and 2B) was conducted batch-wise, as outlined in Table S2. Review of data from Run 1 indicated that drift was occurring based on sample preparation order due to evaporation of solvents with high partitioning to headspace. This also meant that spikes for the accuracy and recovery exercise did not retain integrity compared to the calibration set. To mitigate this effect, an important change was made for Run 2: the same container of standard stock solution was not sampled for all calibrations and spikes (although it had been capped at each step for Run 1). In Run 2, the stock solution was decanted into smaller capped containers and a new container was used for each step in the preparation process. An exception was made for the calibrations, where the same container was used to enable any evaporative losses to be monitored. In addition to differential preparation of calibration standards, they were run in triplicate at the beginning and end of the sequence to gauge possible instrument/sample drift in the method itself. These changes significantly improved performance in Run 2.
For both GC-FID and SIFT-MS measurements, the samples were analyzed in the same sequence for each run, as summarized in Table S2. Different sequences were used for Runs 1 and 2 to reduce drift and better characterize what remained.

Results
This section summarizes linearity, repeatability, accuracy, and recovery results obtained from side-by-side analysis of Class 2A and 2B solvents in water-soluble articles using GC-FID and SIFT-MS. This differs from the previous study [12], in which the specificity, range, and limit of quantitation (LOQ) were also evaluated for SIFT-MS analysis of

Sample Preparation and Analysis
The study was conducted in two separate runs (Table S2) and different commercial standards were utilized for each. For Run 1, USP Class 2A and 2B standards were used (part numbers 1601281 and 1601292, respectively; USP, Rockville, MD, USA), while for Run 2, Sigma-Aldrich Class 2A and 2B standards were employed (part numbers PHR1064 and PHR1065, respectively; Sigma-Aldrich, Gillingham, UK).
Two acetaminophen (paracetamol) formulations were selected for the accuracy and recovery exercise: tablets (500 mg per tablet; Co-op private label, UK) and an oral suspension (250 mg per 5 mL dose; Calpol Six-plus sugar-free, Johnson & Johnson, High Wycombe, UK).
Preparation of samples (combining Classes 2A and 2B) was conducted batch-wise, as outlined in Table S2. Review of data from Run 1 indicated that drift was occurring based on sample preparation order due to evaporation of solvents with high partitioning to headspace. This also meant that spikes for the accuracy and recovery exercise did not retain integrity compared to the calibration set. To mitigate this effect, an important change was made for Run 2: the same container of standard stock solution was not sampled for all calibrations and spikes (although it had been capped at each step for Run 1). In Run 2, the stock solution was decanted into smaller capped containers and a new container was used for each step in the preparation process. An exception was made for the calibrations, where the same container was used to enable any evaporative losses to be monitored. In addition to differential preparation of calibration standards, they were run in triplicate at the beginning and end of the sequence to gauge possible instrument/sample drift in the method itself. These changes significantly improved performance in Run 2.
For both GC-FID and SIFT-MS measurements, the samples were analyzed in the same sequence for each run, as summarized in Table S2. Different sequences were used for Runs 1 and 2 to reduce drift and better characterize what remained.

Results
This section summarizes linearity, repeatability, accuracy, and recovery results obtained from side-by-side analysis of Class 2A and 2B solvents in water-soluble articles using GC-FID and SIFT-MS. This differs from the previous study [12], in which the specificity, range, and limit of quantitation (LOQ) were also evaluated for SIFT-MS analysis of Class 2 residual solvents. Full data for the present study are provided in the Supplementary Materials.

Linearity
Selected linearity data (acquired in Run 1) for SIFT-MS and GC-FID over the range 0.2 to 1.0 of the normalized concentration relative to the solvent limit defined in USP<467> [2] are shown in Figure 5. This is referred to as the "standard level". Full data are given in Tables S3-S6 and Figures S1-S4 for both techniques. Class 2A and 2B results are tabulated separately. SIFT-MS responses are plotted in part-per-million by volume (ppmV) because volume:volume units are readily calculated by software based on instrument, library, and sample measurement parameters [11]. This unit also has the benefit of internally normalizing reagent ion drift (should it occur), since it is proportional to the ratio of product ion to reagent ion during sample measurement [11]. GC-FID responses are shown as peak area. Note that for most analytes with low response (Tables S5 and S6), the upper standard concentration (1.0) in GC-FID data was eliminated for calculation of linear regression coefficients (R 2 ) because a dip in signal was evident. SIFT-MS linear regression coefficients were calculated across the full range (0.2 to 1.0). The results are summarized in Table 1, showing that both techniques easily meet acceptance criteria. SIFT-MS, however, usually performed slightly better.
AppliedChem 2023, 3, FOR PEER REVIEW 8 Class 2 residual solvents. Full data for the present study are provided in the Supplementary Materials.

Linearity
Selected linearity data (acquired in Run 1) for SIFT-MS and GC-FID over the range 0.2 to 1.0 of the normalized concentration relative to the solvent limit defined in USP<467> [2] are shown in Figure 5. This is referred to as the "standard level". Full data are given in Tables S3-S6 and Figures S1-S4 for both techniques. Class 2A and 2B results are tabulated separately. SIFT-MS responses are plotted in part-per-million by volume (ppmV) because volume:volume units are readily calculated by software based on instrument, library, and sample measurement parameters [11]. This unit also has the benefit of internally normalizing reagent ion drift (should it occur), since it is proportional to the ratio of product ion to reagent ion during sample measurement [11]. GC-FID responses are shown as peak area. Note that for most analytes with low response (Tables S5 and S6), the upper standard concentration (1.0) in GC-FID data was eliminated for calculation of linear regression coefficients (R 2 ) because a dip in signal was evident. SIFT-MS linear regression coefficients were calculated across the full range (0.2 to 1.0). The results are summarized in Table 1, showing that both techniques easily meet acceptance criteria. SIFT-MS, however, usually performed slightly better.   Table 1 summarizes the results obtained for repeatability across six replicates in Run 1. Full data are provided in Tables S7a and S8a for SIFT-MS and Tables S9a and S10a for GC-FID for Class 2A and 2B, respectively. For both SIFT-MS and GC-FID, acceptance criteria are met for all compounds. It must, however, be noted that the largest relative standard deviation (RSD) for GC-FID arises only because individual xylene isomers have been summed to enable direct comparison with SIFT-MS. The individual isomers perform better, with m-, o-, and p-xylene having RSDs of 6.9, 4.1, and 6.9%, respectively. Taking this into account, 1,2-dimethoxymethane has the highest RSD (10.5%) for an individual analyte.

Precision
Repeatability data for Run 2 are shown in Tables S7b-S10b of the Supplementary Materials because they emphasize one of the largest challenges of residual solvents analysis: loss of volatile solvents from aqueous standards. In these tables, triplicate standards were analyzed before and after collection of accuracy and recovery data (Table S2). Drift is generally greater for more hydrophobic compounds. Calculation of repeatability from the two sets of triplicate measurements results in larger RSDs than the dedicated repeatability study during Run 1 ( Table 1), but the results are useful for demonstrating that great care needs to be taken with experimental design.
Intermediate precision (SIFT-MS) was not re-investigated here, since it was described previously for different analysts using different standards and samples [12]. The reproducibility of SIFT-MS data across multiple geographically separated instruments and analysts will be described in a future publication.

Accuracy
Accuracy was calculated from the value measured for the spiked sample (0.5 (50%) or 1.0 (100%)) divided by the mean measurement of the 1.0 (100%) level calibrations analyzed in triplicate before and after the run, as shown in Equation (1).
These calibrations are summarized in Tables S7b-S10b for Run 2. During data analysis, it was discovered that several replicates for GC-FID analysis appeared to be compromised because the responses for all analytes were significantly lower than the other two replicates. These are indicated in gray text in Tables S13b, S14b, S17b and S18b, and are not included in calculation of accuracy. Several replicates appear slightly lower in the SIFT-MS data, but not to the same extent as for GC-FID, so these have been retained in all calculations.
Accuracy data from Run 2 (i.e., after improvement of the experimental procedure) are summarized in Table 1 and are calculated from the mean of replicate samples. Figure 6 visualizes the results across individual valid replicates. Full data across both tablet and oral suspension formulations are provided in the Supplementary Materials (Tables S11-S18) for both runs (Runs 1 and 2 have "a" and "b" identifiers, respectively, added to the above numbers).
Overall, the SIFT-MS results display higher accuracy than the GC-FID results, as measured in terms of the number of compounds that meet the acceptance criteria. For the tablet samples, SIFT-MS meets the acceptance criteria for all analytes (within 20% of standard) when means are used (Table 1), although for the occasional replicate, failure is evident (Figure 6). For GC-FID analysis, several failures per spike level occurred. The oral suspension was more challenging for both techniques. The SIFT-MS results will be discussed in more detail below. Just hexane failed at the 0.5 level and one quantitation ion of 1,4-dioxane failed at the 1.0 level.
In summary, GC-FID fails to achieve acceptance criteria more frequently than SIFT-MS. 1,2-Dimethoxyethane is problematic at both spike levels for GC-FID, while various compounds fall just short of the accuracy criteria at one level (Table 1).

Recovery
Analogous to accuracy (Section 3.3), recovery data are summarized here from Run 2, while full data for both runs are given in Tables S19-S26 of the Supplementary Materials.
Recoveries (as a percentage) were calculated against the 1.0-level calibration data (mean across the pre-and post-run calibration samples; Tables S7b-S10b) scaled by the spike level of the sample (0.5 or 1.0; see Equation (2)). The results obtained for SIFT-MS and GC-FID are given in Table 1  Overall, the SIFT-MS results display higher accuracy than the GC-FID results, as measured in terms of the number of compounds that meet the acceptance criteria. For the tablet samples, SIFT-MS meets the acceptance criteria for all analytes (within 20% of standard) when means are used (Table 1), although for the occasional replicate, failure is evident ( Figure 6). For GC-FID analysis, several failures per spike level occurred. The oral suspension was more challenging for both techniques. The SIFT-MS results will be discussed in more detail below. Just hexane failed at the 0.5 level and one quantitation ion of 1,4-dioxane failed at the 1.0 level.
In summary, GC-FID fails to achieve acceptance criteria more frequently than SIFT-MS. 1,2-Dimethoxyethane is problematic at both spike levels for GC-FID, while various compounds fall just short of the accuracy criteria at one level (Table 1).

Recovery
Analogous to accuracy (Section 3.3), recovery data are summarized here from Run 2, while full data for both runs are given in Tables S19-S26 of the Supplementary Materials. Recoveries (as a percentage) were calculated against the 1.0-level calibration data (mean across the pre-and post-run calibration samples; Tables S7b-S10b) scaled by the spike level of the sample (0.5 or 1.0; see Equation (2)). The results obtained for SIFT-MS and GC-FID are given in Table 1 and Figure 7. Failures to meet acceptance criteria largely parallel those described above for accuracy.

Discussion
Analysis of Class 2 residual solvents in the headspace of water according to the USP General Chapter <467> [2] is challenging for several reasons. In physicochemical terms, the solvents vary significantly in volatility and headspace partitioning. They also vary in terms of toxicity, which translates to widely differing permitted daily exposures (PDEs) from 0.5 to 38.8 mg per day [1,2]. For SIFT-MS, with its wide dynamic range and consistent high sensitivity to a broad range of solvents (VOCs) [11,14], the primary challenge is en-

Discussion
Analysis of Class 2 residual solvents in the headspace of water according to the USP General Chapter <467> [2] is challenging for several reasons. In physicochemical terms, AppliedChem 2023, 3 299 the solvents vary significantly in volatility and headspace partitioning. They also vary in terms of toxicity, which translates to widely differing permitted daily exposures (PDEs) from 0.5 to 38.8 mg per day [1,2]. For SIFT-MS, with its wide dynamic range and consistent high sensitivity to a broad range of solvents (VOCs) [11,14], the primary challenge is ensuring that integrity is maintained during sample preparation (ensuring that volatile and hydrophobic solvents are not lost). By modifying our sample preparation approach in a way that minimized the open-close cycles for the 100 mL stock solutions (by creating subsamples), accuracy and recovery results were significantly improved in Run 2 compared to Run 1, resulting in most analytes meeting acceptance criteria. Accurate, precise, and rapid liquid transfers are clearly essential to the reliable performance of this method (c.f., Ref. [4]).
For GC-FID, the lower sensitivity to compounds such as 1,2-dimethoxyethane, 1,4-dioxane, and pyridine may contribute to poorer accuracy and recovery performance than SIFT-MS. Another potential contributing factor that should be explored in a future study is whether the extended run time of the GC-FID batch contributes to analyte losses due to prepared samples sitting for extended periods in the autosampler tray while awaiting analysis (up to 36 h for GC vs. 3.2 h for SIFT-MS; see Figure 2). Losses due to very small leaks in septa, for example, would result in poorer accuracy and recovery. Finally, it should be noted that this study is actually broader than routine USP <467> analysis, so increased failures may be due to operating outside the optimized range of the GC-FID method.
The accuracy and recovery data in Table 1 and Figures 6 and 7 show that SIFT-MS is struggling with analysis of 1,4-dioxane and hexane in the oral suspension. For 1,4-dioxane, the H 3 O + quantitation ion (m/z 89; Table S1) suffers an unidentified interference giving poor accuracy and recovery, whereas the NO + quantitation ions (m/z 87 and 88) do not (Table S15b). Interestingly, in Run 1 the NO + 88 ion also suffered interference for this formulation (perhaps due to trace levels of acetone [27]), but NO + 87 still performed acceptably (Table S15a). This illustrates the utility of multiple, rapidly switchable reagent ions-with orthogonal ionization chemistry-that are applied in SIFT-MS instruments. When one quantitation ion suffers interference, another quantitation ion(s) can often be utilized. In fact, an instrument software feature for interference rejection is ordinarily applied automatically in data processing, giving a single concentration reading (or pass/fail) for each analyte [11]. However, multiple ions are not always an option: hexane is quantified via a single quantitation ion (m/z 85 with NO + ; Table S1). In the oral suspension data, hexane suffers matrix interference at the 0.5 level, causing it to fail in terms of accuracy and recovery. The level of the interference is modest, since acceptance criteria are met at level 1.0. Hence, trace quantitation of hexane may be challenging in this matrix using the standard approach, though the method of standard additions might provide an alternative [32]. When applying the SIFT-MS procedure to new products, use of full-scan analysis across all available reagent ions to identify potential matrix effects is recommended. In routine use, a test failure can be investigated using full-scan SIFT-MS analysis. If this does not confirm the identity of the solvent or identify an interferent, then further investigation should be conducted using conventional chromatographic methods with MS detection.
A more subtle interference issue is evident in some of the SIFT-MS linearity data presented in full in Figures S1 and S2, where there is a lack of agreement between different quantitation ions. This issue is most evident for some isotopologue peaks of 35 132 and 132). The use of multiple product ions reduces the impact of these interferences in the present results, for the reason given in the preceding paragraph. Furthermore, the practical application of the analytical method will ordinarily focus on solvents that are likely to be present (LTBP) per the original risk assessment made by the product manufacturer [2]. Hence, this "cross talk" can be eliminated through use of an appropriate subset of analytes and calibration of individual solvents or use of less complex calibration mixtures. Of course, such decisions should be made product by product during method development and validation-i.e., when the SIFT-MS procedure is tested on a new product, with supporting full-scan analysis.
The present study is clearly of limited scope in that it has only compared GC-FID and SIFT-MS performance for water-soluble articles and hence does not broadly address industry requirements [2,4]. An upcoming publication will establish the concentration ranges over which six compatible non-aqueous diluent solvents can be used with SIFT-MS and thus support extending the applicability of SIFT-MS to non-aqueous and mixed aqueous/organic solvent systems. This evaluation is necessary because headspace is introduced continuously into the SIFT-MS instrument and all solvent present in the gas phase is potentially reactive with the reagent ions.
Despite these challenges, SIFT-MS potentially provides a significant step forward for residual solvents analysis in water-soluble matrices. Applying the benefit of faster sample analysis (Figures 2-4) to a real-world testing scenario that includes daily calibration, it is evident that a SIFT-MS instrument provides significantly faster time to results, as well as much higher sample throughputs ( Figure 8). Additionally, the cost of ownership is reduced through elimination of the chromatographic column (and related maintenance and replacement). For pharmaceutical companies investing in scale-up of their manufacturing processes, the increased analytical capacity achievable with a single automated SIFT-MS instrument may provide significant economic benefits as a high-throughput screening tool for water-soluble articles due to a single SIFT-MS instrument replacing multiple GC-FID instruments.

Conclusions
This study compared the performance of a SIFT-MS-based alternative procedure [1 with the standard USP <467> [2] GC-FID procedure for analysis of Class 2A and 2B resi ual solvents. Both techniques comfortably met the acceptance criteria for linearity an precision (repeatability). However, SIFT-MS outperformed GC-FID for accuracy and r covery measurements, with just one compound failing for SIFT-MS versus ten for GC-FI It is postulated that this is due to FID signal levels and significantly longer wait times f GC-FID analysis compared to SIFT-MS (see Figure 2), which may result in losses of vo tiles and hence increased failures. In addition to reducing the wait time (five-fold fast time to deliver the first analytical result), SIFT-MS can provide operators with a daily sam ple throughput 15-fold greater than GC-FID, delivering lower cost per sample.
Although SIFT-MS is a DIMS technique-that is, it analyzes headspace directly wit out pre-separation using chromatography-it provides specific analysis through applic tion of multiple orthogonal reagent ions [10,11]. Where possible, SIFT-MS methods u several quantitation ions and software cross-compares these independent measuremen to overcome matrix effects. However, at times, matrix complexity and/or reaction chem try can be a limiting factor (here, hexane could not be analyzed effectively in the oral su pension). As part of best-practice application of a SIFT-MS analytical method to a ne ingredient or formulation, the matrix should be assessed using full scan analysis in add tion to targeted analysis of residual solvents. With appropriate method development an validation, SIFT-MS could provide a useful alternative to GC-based approaches for resi ual solvent analysis, especially in environments where fast test results and/or high samp throughputs are required.
Supplementary Materials: The following supporting information can be downloaded

Conclusions
This study compared the performance of a SIFT-MS-based alternative procedure [12] with the standard USP <467> [2] GC-FID procedure for analysis of Class 2A and 2B residual solvents. Both techniques comfortably met the acceptance criteria for linearity and precision (repeatability). However, SIFT-MS outperformed GC-FID for accuracy and recovery measurements, with just one compound failing for SIFT-MS versus ten for GC-FID. It is postulated that this is due to FID signal levels and significantly longer wait times for GC-FID analysis compared to SIFT-MS (see Figure 2), which may result in losses of volatiles and hence increased failures. In addition to reducing the wait time (five-fold faster time to deliver the first analytical result), SIFT-MS can provide operators with a daily sample throughput 15-fold greater than GC-FID, delivering lower cost per sample.
Although SIFT-MS is a DIMS technique-that is, it analyzes headspace directly without pre-separation using chromatography-it provides specific analysis through application of multiple orthogonal reagent ions [10,11]. Where possible, SIFT-MS methods use several quantitation ions and software cross-compares these independent measurements to overcome matrix effects. However, at times, matrix complexity and/or reaction chemistry can be a limiting factor (here, hexane could not be analyzed effectively in the oral suspension). As part of best-practice application of a SIFT-MS analytical method to a new ingredient or formulation, the matrix should be assessed using full scan analysis in addition to targeted analysis of residual solvents. With appropriate method development and validation, SIFT-MS could provide a useful alternative to GC-based approaches for residual solvent analysis, especially in environments where fast test results and/or high sample throughputs are required.