THz bandwidth photonic radio frequency spectrum analyzer based on a CMOS-compatible high-index doped silica waveguide

YUHUA LI City University of Hong Kong ZHE KANG City University of Hong Kong KUN ZHU City University of Hong Kong SHIQI AI City University of Hong Kong XIANG WANG QXP Technology ROY R. DAVIDSON QXP Technology YAN WU Zhejiang Sci-Tech University ROBERTO MORANDOTTI INRS BRENT E. LITTLE Xi’an Institute of Optics and Precision Mechanics David Moss (  dmoss@swin.edu.au ) swinburne university https://orcid.org/0000-0001-5195-1744 SAI TAK CHU City University of Hong Kong

Here, we demonstrate a high-index doped silica based PC-RFSA with improved performance by designing a waveguide structure with low dispersion slope and optimized dispersion for the C-band operation having ZDW at near 1547 nm. We obtained a bandwidth of 5 THz that covers the entire C-band using a 50-cm long waveguide. With the optimized waveguide dispersion, one can obtain broadband operation even with a longer waveguide and results in a higher signal sensitivity. Figure 1 shows the principle of operation of the PC-RFSA. An optical signal under test (SUT) having an RF spectrum described by and a temporal intensity power spectrum given by is mixed with a continuouswave (CW) probe beam, with electric eld given by ), and both are then launched into a nonlinear medium [4]. Due to the near-instantaneous response of the cross-phase modulation (XPM) effect [19] in the medium, the phase of the CW probe beam is modulated by the intensity of the SUT, with a modulation index of , where and are the nonlinear parameter and effective propagation length of the waveguide, respectively. As the combined signals propagate through the nonlinear waveguide, frequency modulation sidebands are generated around , with their combined optical spectrum being proportional to due to the XPM [4]. This results in the RF spectrum of the SUT being mapped onto the sideband of the optical spectrum of the probe light, which can then be measured directly by an optical spectrum analyzer (OSA).
Since this is an all-optical operation, in principle it can achieve unlimited bandwidth as it is not limited by the electronic bandwidth from optical-to-electrical conversion. In practice, effects such as group velocity mismatch between the SUT and the CW probe light, limit the bandwidth according to where D is the dispersion of the medium, ∂D/∂λ is the dispersion slope at the wavelength where the SUT is centered, and Δλ is the wavelength spacing between the SUT and the CW probe light [11]. Therefore, waveguides with both low dispersion and low dispersion slope are crucial to maximize the device bandwidth. The optimal bandwidth for a given probe to SUT wavelength separation of Δλ is obtained by setting the probe and SUT wavelength symmetrically with respect to the ZDW [4]. For C-band application, the ideal ZDW location should be at the center of the band of around 1547 nm.
To optimize the waveguide for the PC-RFSA for operation in the C-band, we calculate the waveguide dispersion as a function of geometry and wavelength. show the calculated nonlinear parameter for the TE and TM modes of the waveguide versus thickness. The nonlinearity increases with decreasing thickness and wavelength due to stronger optical con nement. However, at 1547 nm there is minimal change in for thicknesses from 0.5 mm to 1.5 mm.
A series of 50-cm and 100-cm long spiral waveguides with a width of 2 μm and thickness ranging from 1.00 μm to 1.50 μm in steps of 0.25 μm were fabricated using the process described in [20][21][22][23][24][25][26] to investigate their performance as PC-RFSAs. The inset in Fig. 3 shows a scanning electron microscope (SEM) image of the fabricated buried waveguide with cross-section 2 μm × 1.25 μm. The core refractive index is 1.7 and the cladding index is 1.45. It can be seen that there is a sidewall angle of approximately 8° resulting from the etching process. With the mode transformers at the input and output ends of the spirals, the coupling loss between the optical ber and waveguide is about 1 dB/facet, while the propagation loss varies from 0.06 to 0.1 dB/cm at 1547 nm for the different thicknesses. The calculated dispersion curve for a 2 μm 1.25 μm waveguide (Fig. 3) shows that the ZDWs are at 1542 nm and 1503 nm for the TE and TM modes, respectively. At 1547 nm, the desired operation point for C-band application, the dispersion is −2.6 ps•nm −1 •km −1 for the TE mode and −22.1 ps•nm −1 •km −1 for the TM mode, while the dispersion slopes are −0.47 ps•nm −2 •km −1 and −0.55 ps•nm −2 •km −1 for TE and TM modes, respectively. The dispersion for the TE mode across the whole C and L bands remains low at < 50 The setup for the PC-RFSA measurement is shown in Fig. 1 where tunable narrow-linewidth laser sources, TLs1 and TLs2, with wavelengths at and , respectively, served as the SUT and TLs3 with wavelength as the CW probe. The two SUT signals were initially set at 1547 nm to fully utilize the entire C-band spectrum. Their spacings were then detuned in opposite directions from the C-band center to simulate the broadening of the SUT frequency spectrum in the measurement. The probe signal from TLs3 was xed at 1601 nm, far away from the SUT signals to avoid spectral overlap between the four-wave-mixing signal from the SUT and XPM signal from the probe, when the bandwidth measurement is performed in the whole C-band. PCs, PBS and PM were used to ensure that the polarization states of all the three signals were aligned to the TE or TM mode of the waveguide. High-index doped silica 4 2.5 [11] High-index doped silica 50 5.0 This work Figure 4 shows the normalized XPM power as a function of frequency separation between TLs1 and TLs2 for the set of 50-cm spiral waveguides having thicknesses between 1.00 mm to 1.50 mm for the TE mode with each of the measured data tted to parabolic curves. For the 1.25 μm thick waveguide, we obtain a 3-dB bandwidth for the TE mode of ~ 5 THz and 3.43 THz for the 50 cm and 100 cm waveguides, respectively. Table 1 summarizes the results from the literature, showing that the 5 THz bandwidth obtained in this work corresponds to the highest value reported so far. We also obtained a XPM walk-off delay, de ned as [8] of 70 for the 1.25 mm thick waveguide, indicating the walk-off delay in uence on XPM is extremely small and the nonlinear interaction is maintained over large ∆λ. Table 2 lists the measured 3-dB bandwidths for all three fabricated waveguides using the experimental setup in Fig. 1 and the same wavelengths. Since the ZDWs are located outside of the C-band for the 1.00 mm and 1.50 mm waveguides, their obtained bandwidths are limited. However, one can shorten the waveguide to further increase the bandwidth if desired, similar to [11].  Figure 5 compares the PC-RFSA sensitivity for a 50-cm and a 100-cm spiral waveguide, with the sensitivity de ned as the power level of the generated XPM signal that is 3 dB above the OSA (Yokogawa, AQ6370D) noise oor [2] for a given probe power. This represents the minimum power level that the PC-RFSA can detect the 3-dB bandwidth of the SUT. The frequency separation between the SUT signals was set to 625 GHz, with λ1, λ2, λ3 set at 1550 nm, 1555 nm, 1580 nm, respectively. By setting the probe power to 0 dBm, the relationship between the input SUT power and generated XPM power from the 50-cm and 100-cm long spirals is shown in Fig. 5. Due to the accumulated nonlinear effect and low propagation loss of the waveguide in the C-band, the 100-cm long spiral waveguide produced a higher XPM signal. At the lowest noise oor setting of the OSA at −88 dBm, the minimum resolvable input SUT power value was 1.5 dBm for the 100-cm long waveguide, which was 2 dB lower than that for the 50-cm long waveguide.
To investigate the distortion of the PC-RFSA, we measured the RF spectrum of a femtosecond ber laser the temporal intensity from autocorrelation measurements with its corresponding optical spectrum. The RF spectrum of the femtosecond pulse, measured by the PC-RFSA, is depicted in Fig. 6(c). The PC-RFSA measured the RF spectrum as broad as 2 THz. The dashed curve in Fig. 6(c) represents the simulated RF spectrum, t to a sech 4 function [9]. As shown in Fig. 6(c), the measured RF spectrum agrees well with theory, which proves that our PC-RFSA is suitable for RF spectrum analysis of signals with bandwidths in the terahertz regime. These results further add to the substantial body of work on microwave signal generation and processing based on this nonlinear optical platform  In conclusion, we present a PC-RFSA that achieves an RF bandwidth extending well into the terahertz regime, based on a high-index doped silica waveguide. We investigate the dispersion and nonlinear parameter for different waveguide dimensions. By engineering the dispersion, a 3-dB bandwidth of 5 THz is achieved, which is twice as large as that previously reported [11]. The measurement of the RF spectrum of a femtosecond ber laser demonstrates its usefulness for frequency spectrum analysis of ultrabroadband optical signals involving frequencies in the terahertz range.

Declarations
Funding. The Natural Science Foundation of Zhejiang Province (LY21F050007); Zhejiang Provincial Natural Science Foundation of China (LGG21A040003); The National Natural Science Foundation of China (62075188).
Disclosures. The authors declare no con icts of interest   Normalized XPM power as a function of the frequency spacing between TLs1 and TLs2 for TE mode. The measured data are t with the 2nd order polynomial function.

Figure 5
Page 15/15 The XPM sideband power as a function of the averaged input SUT power for the 1.25 μm thick waveguides and different lengths (TE mode is considered and the SUT frequency is xed on 625 GHz).