Determination of Chromium in Natural Water by Adsorptive Stripping Voltammetry Using In Situ Bismuth Film Electrode

Development of adsorptive stripping voltammetry (AdSV) combined with in situ prepared bismuth film electrode (in situ BiFE) on glassy carbon disk surface using diethylenetriamine pentaacetic acid (DTPA) as a complexing agent and NO3− as a catalyst to determine the trace amount of chromium (VI) is demonstrated. According to this method, in the preconcentration step at Edep = −800 mV, the bismuth film is coated on the surface of glassy carbon electrodes simultaneously with the adsorption of complexes Cr(III)-DTPA. In addition to the influencing factors, the stripping voltammetry performance factors such as deposition potential, deposition time, equilibration time, cleaning potential, cleaning time, and technical parameters of differential pulse and square wave voltammetries have been investigated, and the influence of Cr(III), Co(II), Ni(II), Ca(II), Fe(III), SO42−, Cl−, and Triton X has also been investigated. This method gained good repeatability with RSD <4% (n = 9) for the differential pulse adsorptive stripping voltammetry (DP-AdSV) and RSD < 3% (n = 7) for the square wave adsorptive stripping voltammetry (SqW-AdSV), and low limit of detection: LOD = 12.10−9 M ≈ 0.6 ppb (at a deposition potential (Edep) of −800 mV and the deposition time (tdep) of 50 s) and LOD = 2.10−9 M ≈ 0.1 ppb (at Edep = −800 mV and tdep = 160 s) for the DP-AdSV and SqW-AdSV, respectively. This method has been successfully applied to analyze chromium in natural water.


Introduction
In the natural environment, chromium exists in two thermodynamically stable states, Cr(VI) and Cr(III). Its toxicity depends on the level of oxidation; Cr(VI) is the causative agent of cancer through the airways and toxic to humans and other mammals, while Cr(III) at trace levels provides the body with essential minerals. e sources of chromium emissions into the environment are electroplating, metallurgical, dye processing, tanning, minerals, and refractory materials [1,2]. e form of chromium does not only determine its ecological impact but also determines its movement and variation in the environment. Studies related to the transformation of Cr(VI), the distribution of Cr(III) between inorganic and organic compounds, and the treatment of chromium-contaminated environment are increasingly being considered [3,4].
ere are several sensitive methods for determining chromium such as atomic absorption spectrometry [5,6], plasma emission spectrometry [7,8], and capillary gas chromatography [9]. ese methods, while highly sensitive, only reach the ppb size detection limit and often require preenrichment and extraction stages, which can lead to sample contamination, complicate processes, and increase the cost of analysis, and therefore, it will be difficult to require frequent and mass analysis of samples in the environment. To overcome these difficulties, in recent years, analysts have successfully applied the AdSV method. is method allows direct determination (i.e., no need to extract or evaporate the sample) of the amount of traces, even super chromium traces in different objects, and achieves very low LOD, size < 10 −9 -10 −12 M [10][11][12][13][14][15][16][17][18][19][20].
When analyzing Cr(VI) by the AdSV method, Yong et al. [10] and Lin et al. [11] proposed the adsorption mechanism as follows.
e first stage of the adsorptive stripping voltammetry method is the movement of Cr(VI) in the solution to the electrode surface and is reduced by the following reaction.
Cr(VI) + 3e ⟶ Cr(III); E 1/2 � −0.2 V in comparison with 3 M Ag/AgCl/KCl electrode; in the presence of DTPA ions on the interface between the electrode membrane and the solution, Cr(III) ions form complexes quickly and adsorbed onto the electrode surface. e main nature of this complex is unknown, and some authors [21,46] said that this complex exists mainly as Cr(III)-DTPA 2− and a little in Cr(III)-HDTPA − , or Khan [47] suggested that the complex of Cr(III) with DTPA exists as [Cr(III) (H 2 O)HY] − , [Cr(III)-DTPA] 2− , and so on.
After complex adsorption, potential scan was conducted from −0.9 V to −1.35 V [10], or from −0.8 V to −1.4 V [11]; as a result of this process, the Cr(III)-DTPA complex is reduced to the Cr(II)-DTPA complex [10,11]: e signal recorded with this reduction is at E 1/2 � −1.15 V [10,11] and at E 1/2 � −1.1 V in comparison with 3 M Ag/AgCl/ KCl electrode. e role of NO − 3 ions is to oxidize Cr(II)-DTPA complexes into Cr(III)-DTPA complexes.
anks to the oxidation of NO − 3 , the peak is higher and the sensitivity of the method is increased.
In order to contribute to the development of the AdSV method, we have conducted several studies to determine chromium by the AdSV method using mercury film electrodes, ex situ bismuth film electrode, and in situ bismuth film electrode. In this paper, we present the results of chromium determination by the AdSV method using in situ bismuth film electrode (in situ BiFE) in the presence of DTPA as a complexing agent and ion NO − 3 as a catalyst. Literature survey revealed that in situ BiFE has never been used for chromium determination by the adsorptive stripping voltammetry.

Apparatus and Reagents.
Square wave stripping voltammetric measurements were conducted on Electrochemical Analyzer 797 VA Computrace (Metrohm, Switzerland) accompanied with three electrodes. ese electrodes were inserted into the 80 ml capacity electrochemical cell. e working electrode was a glassy carbon rotating disk electrode with d � 2.8 ± 0.1 mm; the reference electrode was a 3 M Ag/AgCl/ KCl electrode and the auxiliary electrode was a platinum wire. All measurements were carried out at 25 ± 1°C. DTPA (diethylenetriamine pentaacetic acid) is used as a complexing agent for chromium. 50 Other metal ionic solutions such as Co(II), Ni(II), Zn(II), Cr(III), Fe(III), and Ca(II) are from the corresponding 1000 mg/l stock solution (the type used for analyzing atomic absorption spectrometry from Merck). Triton X-100 working solution was prepared from Triton X-100 (Merck). Journal of Environmental and Public Health 3  Journal of Environmental and Public Health  [10].
At the end of this period, the electrode stops rotating for 30-60 seconds (equilibration time, t equal ). Subsequently, the potential scan was carried out in a negative potential direction from −800 mV to −1450 mV, and at the same time, the stripping voltammogram was recorded using a certain stripping voltammetry technique, differential pulse adsorptive stripping voltammetry (DP-AdSV), or square wave adsorptive stripping voltammetry (SqW-AdSV). During this period, Cr(III) in the Cr(III)-DTPA complexes is reduced to Cr(II) forming Cr(II)-DTPA complexes and generating the stripping peak current of chromium (I p ) [10]. If NO − 3 is not present in solution, I p will be very small, NO − 3 present in the solution will oxidize Cr(II)-DTPA to Cr(III)-DTPA and then Cr(III)-DTPA is electrochemically reduced to Cr(II)-DTPA, and the repeated cycle increases the height of I p [10]. In other words, the NO − 3 ion acts as a catalyst. After dissolving, the electrode was cleaned by electrolysis at +400 mV for 30 seconds to dissolve Bi o and other metals that may be present into the solution. I p is proportional to the concentration of Cr(VI) in the solution.
In all experiments, for Cr(VI) with trace, the first measurement result must be discarded because it is unstable. e stripping voltammogram was recorded 3 times (n � 3), and the peak current (I p ) and peak potential (E p ) values are averaged from three repetitions. e glassy carbon electrodes were cleaned by polishing the surface with fine Al 2 O 3 powder (particle size 0.6 μm) and then washed with distilled water and then with 1 M NaOH to remove all Al 2 O 3 particles on the glassy carbon surface, and then the electrodes were dipped into 1 M HCl solution and finally washed with distilled water and the electrodes were dried with soft filter paper.

Results and Discussion
3.1. Differential Pulse Adsorptive Stripping Voltammetry (DP-AdSV) Using In Situ BiFE. In order to select the appropriate conditions for the method, the experimental conditions were fixed as shown in Table S1. A univariate method is applied to examine the effect of factors. e magnitude of the stripping peak current (I p ) and the relative standard deviation of the I p (RSD) are used for selecting the appropriate test conditions.

Effect of Acetate Buffer Concentration.
Acetate buffer is chosen to stabilize the pH of the solution. Acetate buffer is one of the factors that strongly influence the complex of Cr(III) and DTPA [10]. e complex formation between Cr(III) and DTPA usually occurs at pH � 6 [10,11]. At chromium concentration C Cr(VI) � 3.8.10 −8 M, ligand concentration C DTPA � 0.4.10 −3 M, and bismuth concentration C Bi(III) � 24.10 −5 M, the survey results of the effect of acetate buffer concentration (C Ac ) in the range of 0.1 M-0.6 M (pH � 6) showed that C Ac � 0.4 M was appropriate. With this condition, the peak current is 31.36 μA and the repeatability is relatively good (RSD � 1.6% with n � 3) (Figure 1(a)).

Effect of Bi(III) Concentration and DTPA
Concentration. Previous studies with ex situ BiFE electrodes have suggested that the presence of KBr in the solution increases the bismuth's sustainability on the glassy disk surface and at the same time improves the conductivity of the solution [48]. In the acetate buffer with C Ac � 0.4 M and the presence of KBr at a concentration of 5.0.10 −6 M, C DTPA � 0.4.10 −3 M, and C Cr(VI) � 3.8.10 −8 M, a Bi(III) concentration of 28.8.10 −5 M is appropriate. At those concentrations, the peak current is 37.6 μA and the repeatability is good (RSD � 1.7% with n � 3) (Figure 1(b)). e survey results of the effect of DTPA concentrations in the range of 0.1 to 1.0.10 −3 M to Cr peak current show that DTPA concentration of 0.4.10 −3 M was appropriate (Figure 1(f )).

Effect of NaNO 3 Concentration.
In the presence of NO − 3 , peak current of chromium (I p ) was enhanced significantly. Many authors argue that NO − 3 ions act as oxidizing agents that convert Cr(II)-DTPA complexes to Cr(III)-DTPA complexes and thus increase the concentration of Cr(III) on the electrode surface, and this leads to an increase in I p [10,11,21]. At  (Figures 1(e) and 2(c)). However, when the NaNO 3 concentration is greater than 0.4 M, it increases the baseline and may contaminate the analysis solution. When the NaNO 3 concentration is equal to 0.4 M, the peak current is 17.6 µA and the repeatability is quite good (RSD � 3.7% with n � 2). NaNO 3 concentration value of 0.4 M was selected for further investigation.  Figure 1(c) and the stripping voltammetry is shown in Figure 2(a).

Effect of Deposition Potential (E dep ) and Deposition
From this result, it shows that when E dep is equal to −800 mV, the peak current is 42.1 µA and the repeatability is good (RSD � 1.3% with n � 3). E dep of −800 mV was selected for further studies.
With the above conditions and when E dep is equal to −800 mV, I p is almost unchanged when the deposition time (t dep ) is greater than 80 s, which means that it tends to reach saturation (Figure 1(d)). t dep of 50 s is selected for the next experiment (the stripping voltammetry at t dep � 50 s is shown in Figure 2(b)).

Effect of Rotating Rate of Electrode (ω) and
Equilibration Time (t equal ). By increasing the rotation speed of the  Table S1.
electrode to a specified value, it will increase the mass transfer and the efficiency of the enrichment will be better. e survey results of the rotating rate of the electrode in the range from 800 rpm to 2400 rpm showed that ω of 2000 rpm was appropriate. At the end of the enrichment phase, the electrode should not be rotated for a specified period of time to keep the solution quiet and the electrode surface is stabilized (this time is also called equilibration time, symbolized as t equal ). I p survey results according to t equal showed that t equal of 50 s was appropriate.

Effect of Cleaning Potential (E clean ) and Cleaning Time (t clean ).
e cleaning of the surface of the carbon glassy disc electrode at the end of each stripping voltammetry is essential, as it will create repeating electrode surfaces for subsequent measurements. In terms of experimental conditions as in Section 3.1.3, the survey results of the influence of E clean in the range of 200 mV to 500 mV and t clean in the range of 60 s to 120 s showed that E clean is equal to 300 mV and t clean is equal to 110 s which are appropriate.

Square Wave Adsorptive Stripping Voltammetry (SqW-AdSV) Using
In Situ BiFE. Some authors argue that, in addition to differential pulse stripping voltammetry techniques, square wave stripping voltammetry can be used to record the signal (at this time, the method is called square wave adsorptive stripping voltammetry (SqW-AdSV)) and also allow the determination of very sensitive chromium. Based on the experimental conditions initially fixed as shown in Table S1, the effects of the factors were investigated in a similar way to the DP-AdSV method, and we obtained the appropriate conditions for SqW-AdSV (using in situ BiFE) to determine Cr(VI) as shown in Table S2.

Interferences.
Interferences for the determination of Cr(VI) consist of metallic cations that have the stripping peak current near the stripping peak current of chromium and anions that can form complexes or make conjugates with the forms of chromium and Bi(III) which can be adsorbed onto the surface of the in situ BiFE, and surfactants can be adsorbed onto the working surface of the electrode. e influence of interferences can be estimated by relative error values of stripping peak current (RE). Consider that RE for I p was equal to RE for C (because I p � kC). RE for I p (or C) was accepted when it was equal to ½ Horwitz function RSD (RE I p(Cr) ≤ ½ RSD Horwitz � ½.2 (1−0.5lgC) � 32% with C � 0.2 ppb). RE was calculated as follows: where RE is the relative error values of stripping peak current, I p (Cr) 0 is the stripping peak current without adding interferences and I p (Cr) is the stripping peak current with adding interferences.

Interference Studies.
When the chromium(III) concentration is about 100 times higher than the chromium(VI) concentration, the I p does not change significantly (RE < 18%). In fact, rarely encountered C Cr(III) case is 300 times higher than C Cr(VI) , so it can be assumed that Cr(III) does not affect the Cr(VI) determination. is investigation again confirms that Cr(III) does not affect the determination of Cr(VI) (Table S3 and Figure S3). In the acetate buffer (pH � 5-6), Zn(II), Co(II), and Ni(II) can affect the determination of Cr(VI) because it has a stripping peak current close to the stripping peak current of Cr(VI). E p (Zn) ≈ −1040 ÷ −1050 mV, E p (Co) ≈ − 1290 ÷ −1300 mV, E p (Ni) ≈ −1080 ÷ −1100 mV, and E p (Cr) ≈ −1180 ÷ −1240 mV.
From the experimental results at approximately 3.8 10 −9 M (0.2 ppb) chromium concentration, 120 s deposition time, and the suitable conditions as in Table 2, we have seen that Zn has not influenced the determination of chromium when Zn(II) concentration is 800 times larger than Cr(VI) concentration (RE Ip(Cr) ≤ 16%) ( Table 2), and in consequence, we can determine chromium in the natural water with attendance of Zn(II) because ordinarily Zn(II) concentration is 500 times smaller than Cr(VI) concentration in natural water (Table 2). Co and Ni have not influenced the determination of chromium when Co(II) and Ni(II) concentrations are 90 times larger than Cr(VI) concentration (RE I p(Cr) � 1.7-8.0% for Co(II) and 3.9-19.4% for Ni(II) ( Table 2).
In natural water, Fe(III) and Ca(II) usually exist at high concentrations of mM; in seawater, Ca(II) exists at quite high concentration, about 10 −2 ÷ 10 −3 M, so it is necessary to examine the effect of Fe(III) and Ca(II) on the stripping peak current of Cr(VI). e results in Table S4 show that when the concentration of Fe(III) and Ca(II) increases to 36.10 −6 M and 50.10 −6 M, respectively, meaning that the concentrations of Fe(III) and Ca(II) are about 10,000 times greater than the concentration of Cr(VI), the determination of Cr(VI) is not affected. RE is less than 17%.
In natural water, Cl − and SO 2− 4 ions have significant concentrations (mM and larger), and they can form complexes with metals present in the study solution and may therefore affect the determination of Cr(VI). To investigate the effects of Cl − and SO 2− 4 , a series of experiments were conducted with Cl − concentrations ranging from 0 to 281.7.10 −3 M and SO 2− 4 concentrations ranging from 0 to 10.4.10 −3 M. e results in Table S5 show that Cl − does not affect the determination of Cr(VI) when C Cl− is in the range of 0 to 14.09. 10 −3 M. When C Cl− is greater than 28.17.10 −3 M (nearly equivalent to Cl − concentration in brackish water), Cl − affects the determination of Cr(VI) with RE > 32%. erefore, when analyzing Cr(VI) in samples with high Cl − concentration, it is necessary to take Cl − removal method from the sample. SO 2− 4 did not affect the determination of Cr(VI) by SqW-AdSV/in situ BiFE method with RE < 27%.
In the adsorption stripping voltammetry method, the surfactant can be adsorptive on the surface of the working electrode, and this can affect the adsorption process of the metallic complexes on the working electrode. Triton X-100 (polyethylene glycol) is a typical nonionic surfactant and usually is used in order to observe the influence of the surfactant on the adsorption stripping voltammetry method. e effects of Triton X-100 are investigated at concentrations between 0 and 93. 10 −9 M, and the results in Table S6 show that, when increasing the concentration of Triton X-100 to 25 times higher than the concentration of Cr(VI), it still did not affect the determination of Cr(VI) with RE < 8%.
In fact, the concentration of natural surfactants is rarely greater than 77.10 −9 M, and therefore, it can be assumed that they do not affect the Cr(VI) determination. us, when determining Cr(VI) by the adsorption stripping voltammetry method, it is not necessary to remove the surfactant from the analytical solution.
In some cases, natural water and wastewater contain many organic substances including surfactants. It is necessary to treat the sample to exclude organic substances before analysis using UV irradiation and decomposition in acid mixture.

Repeatability.
Repeat recording of 7 stripping voltammetry lines (n � 7) on the same in situ BiFE according to the DP-AdSV or SqW-AdSV method in Figure 3 shows that I p in both approaches has good repeatability with RSD < 4% (n � 9) and RSD < 3% (n � 7), respectively, for DP-AdSV and SqW-AdSV. e stripping peak of chromium (E p ) is negligible, only about 20 mV toward the positive side.

Linear Range and Detection
Limits. e linear range and LOD of the two methods SqW-AdSV and DP-AdSV Journal of Environmental and Public Health were investigated with the appropriate experimental conditions as shown in Table S2 and the stripping voltammetry specifications as shown in Table S1, and the following results were obtained: (1) e Linear Range. For the SqW-AdSV method, I p and C Cr(VI) have a good linear correlation in the range C Cr(VI) � 0.3 ÷ 1.8 ppb with a correlation coefficient (R) of 0. 9994 (linear regression equation is shown in Figure 4(a), and the stripping voltammetry is shown in Figure 4(b)).
For the DP-AdSV method, there is a good linear correlation in the range C Cr(VI) � 2 ÷ 12 ppb with R � 0.9989 (linear regression equation is shown in Figure 4(d)).
(2) Sensitivity. e SqW-AdSV method achieved a sensitivity (23 µA/ppb) of about 34 times higher than the DP-AdSV (0.682 µA/ppb) method. us, the SqW-AdSV method achieves a narrower linear range than the DP-AdSV method, but it achieves higher sensitivity than the DP-AdSV method (due to its lower LOD and greater slope linearity). It can be said that with LOD as above, DP-AdSV and SqW-AdSV methods can be used with in situ BiFE to determine the trace amount of Cr(VI).

Determination of Chromium in Natural Water by the SqW-AdSV Using In Situ BiFE.
In natural water samples, chromium usually exists in both Cr(VI) and Cr(III) forms. As investigated, the SqW-AdSV/in situ BiFE method identifies Cr(VI) and also determines the total Cr(VI) + Cr(III) if during the decomposition of the sample, an additional oxidizer is added to oxidize Cr(III) to Cr(VI).
us, we can determine chromium in individual forms by determining Cr(VI) ( * ) and total   Table S2.
Cr(VI) + (III) ( * * ). It follows that the Cr(III) content is the difference of ( * * ) and ( * ). Based on the above results, it is possible to apply the SqW-AdSV/in situ BiFE to determine the trace of Cr(VI) with LOD ≈ 0.1 ppb. With that LOD, the SqW-AdSV/in situ BiFE can directly determine the amount of Cr(VI) in natural water, without the stage of getting rich, and this is a great advantage of the SqW-AdSV/BiFE method.  Table S2. In order to answer the question of whether the analytical method used to analyze the amount of chromium in natural water samples can be applied, we have conducted experiments to verify the correctness (through the Certified Reference Material (CRM)) and an analysis of some natural water samples. On the basis of the experiments mentioned above, the analysis process of Cr(VI) and total Cr(III, VI) in natural water by the SqW-AdSV method was proposed.

Quality Control of Analytical Methods through Standard Sample Analysis.
In order to confirm the practical applicability of the SqW-AdSV method to analyze chromium traces using BiFE electrodes, it is necessary to control the analytical method quality by evaluating the accuracy and repeatability when analyzing standard samples.
(1) For Surface Water Samples. Surface water Certified Material Reference (SPS-SW1 Batch 122) was selected to evaluate the accuracy of the method. e actual value of the chromium content of the sample is 2.00 ± 0.02 ppb (95% confidence boundary e � ± 0.02 ppb). Analysis of standard SPS-SW1 surface water (CRM) samples by the SqW-AdSV using in situ BiFE with the appropriate experimental conditions is shown in Table 3. e analysis was repeated 3 times. e volume of the solution to be charged to the electrolyser is 2 mL, and the volume of solution in the electrolyser is 10 mL.
e results in Table 4 show that the SqW-AdSV/in situ BiFE has good repeatability (the standard deviation is 4% for the repetition of 3, and it is less than half the standard deviation based on the Horwitz function (RSD H � 2 (1−0,5lgC) ; when the chromium concentration is 2 ppb, the standard deviation of the Horwitz equation is 41%) [49] and the method has good accuracy because the chromium content is within the 95% confidence interval of the CRM sample.
(2) For Seawater Samples. Analysis of the standard seawater CRM coded NASS 6 by the SqW-AdSV/in situ BiFE with the appropriate experimental conditions as shown in Table 3.
Because the concentration of chromium in NASS 6 seawater was too small to be directly analyzed, only NASS 6 standard sample was used as the matrix for analysis and validity. e actual value of the chromium content in the NASS 6 sample is 0.116 ± 0.008 ppb (the 95% confidence bound e � ± 0.008 ppb). NAAS 6 standard sample was added with standard Cr(VI) solution to attain 3 levels of 2 ppb, 6 ppb, and 10 ppb and then analyzed with the standard added samples to determine recovery. e results showed that the SqW-AdSV/in situ BiFE for chromium analysis in seawater samples has a good accuracy (recoverability from 94 to 109%). According to the AOAC (American Association of Analytical Chemists) when analyzing the levels of 1.0 to 10 ppb, achieving a recovery rate of 80 to 110% is acceptable [50]. erefore, it is possible to use this method to analyze chromium in seawater samples (Table 5).
e results of the linear range, sensitivity, limit of detection, and accuracy showed that it is possible to use [Cr(VI)] is the concentration of Cr(VI) in the electrolyte minus the blank. White sample has [Cr(VI)] � 0.034 ppb; C Cr is the Cr content in the sample (calculated by the formula: C Cr � [Cr(VI)] .V 2 /V 1 ). V 1 : volume of solution taken into the electrolyser (V 1 � 2 ml), V 2 : volume of solution in the electrolyser (V 2 � 10 mL), S is the standard deviation. Experimental conditions are as shown in Table 3. Recovery � (x 2 −x 1 ).·100/x 0 ; S is the standard deviation; experimental conditions are as shown in Table 3.
SqW-AdSV/in situ BiFE to determine chromium in surface water and seawater.

Real Sample Analysis.
For the purpose of testing the possibility of applying the SqW-AdSV/in situ BiFE method for the analysis of chromium in water environment, well water, tap water, lagoon water, and seawater in some different areas in ua ien Hue province were taken for analysis. Water samples were taken in clean PET bottles and acidified with concentrated HCl (500 μl HCl/500 ml of sample). Samples were filtered through 0.45 μm porous fiberglass filter paper and analyzed immediately after filtration.
Collected and stored samples were analyzed directly (after filtration through a 0.45 μm porous fiberglass filter paper) to determine the Cr(VI) content by the SqW-AdSV/ in situ BiFE; the total chromium content of the sample was determined after the decomposition of the sample by the method of (a) ((a): add 50 μl of concentrated HCl, 25 μl of H 2 O 2 35% to 50 ml of sample in the Teflon cup, boil for 90 minutes, let it cool, and adjust up to 25 ml) [51].
Samples were analyzed by SqW-AdSV/in situ BiFE according to the process shown in Figure S2. e results of actual sample analysis are presented in Table 6.

Conclusion
Using in situ BiFE electrodes with DTPA complexing ligands in acetate buffer solution pH 6 with the presence of KBr and NO − 3 ion, DP-AdSV and SqW-AdSV methods can determine chromium(VI) concentrations of 0.3 ppb and 2.0 ppb, respectively. e proposed method has been successfully applied for chromium analysis in some natural water samples such as lagoon water, well water, tap water, and saltwater in some areas of ua ien Hue province, Vietnam. is C Cr(VI+III). analysis procedure in water sample by SqW-AdSV/in situ BiFE method was satisfactorily applied for the determination of chromium in real water such as tap water, river water, and well water samples in all countries of the world. e determination of chromium in the above real water sample could be carried out within 60 min.

Data Availability
e data used to support the findings of this study are available from the corresponding author upon request.

Conflicts of Interest
e authors declare that they have no conflicts of interest. Table S1: for the DP-AdSV and SqW-AdSV using in situ BiFE. Table S2: suitable experimental conditions for DP-AdSV and SqW-AdSV methods using in situ BiFE. Table S3: influence of chromium(III). Figure S1: SqW-AdSV/BiFE in situ stripping voltammograms of chromium(VI) when examining the effects of chromium(III).  .  Table S6: influence of Triton X-100. Figure S2: diagram of C Cr (VI + III) analysis procedure in water sample by SqW-AdSV/in situ BiFE method. . (Supplementary Materials)